RE: [Histonet] Reichert Histostat manual
Thank you for replying. Don Birgerson has already contacted me about
the manual and will send a PDF file.
Here's hoping that someone on the list has one of those disposable knife
holders and no longer needs it. Maybe we can make a deal.
Senior Research Technician
Electron Microscopy Core Facility
Department of Otolaryngology
Washington University School of Medicine
660 S. Euclid Ave., Campus Box 8115
St. Louis, MO 63110
From: Gayle Callis [mailto:email@example.com]
Sent: Friday, March 11, 2005 12:51 PM
To: Lett, Jaclynn; Histonet@lists.utsouthwestern.edu
Subject: Re: [Histonet] Reichert Histostat manual
The disposable knife holder that works in this old cryostat is the one
invented for use with an AO 820 microtome many, many years ago. It is a
solid holder with a preset angle that needs no adjustment. You can slip
high profile blade (only profile blade you can use in this holder) using
side release lever. The joy is holder bottom parts fit into the slots
this cryostat, and will lock securely into place. You have to play with
picking up sections a bit but it works.
There may some of these old disposable knife holders floating around
storage rooms and someone willing to give one to you.
Sorry, our manual is long gone but
contact Don.Birgerson@leica-microsystems.com - he is a great source of
info for these old instruments.
One thing, keep all sliding parts well oiled with cryostat oil in this
machine is a must but it will work. That side arm knife holder was a
horrible monster to use, and not the safest design either.
At 08:24 AM 3/11/2005, you wrote:
>We recently inherited a Reichert Histostat 855 made by Cambridge (the
>manufacturer's label also says "975CJ Cryostat Microtome"). Its
>operation manual could not be found. Does anyone have a manual and
>would you be willing to send us a copy? It would be greatly
>Also, does anyone know of a disposable knife holder suitable for this
>model? The steel knife holder in this machine clamps the knife on its
>I hope someone can assist us.
>Thank you so much,
>Senior Research Technician
>Electron Microscopy Core Facility
>Department of Otolaryngology
>Washington University School of Medicine
>660 S. Euclid Ave., Campus Box 8115
>St. Louis, MO 63110
The materials in this message are private and may contain
>Healthcare Information. If you are not the intended recipient, be
>that any unauthorized use, disclosure, copying or the taking of any
>in reliance on the contents of this information is strictly prohibited.
>you have received this email in error, please immediately notify the
>sender via telephone or return mail.
>Histonet mailing list
Research Histopathology Supervisor
Veterinary Molecular Biology
Montana State University - Bozeman
PO Box 173610
Bozeman MT 59717-3610
406 994-6367 (lab with voice mail)
406 994-4303 (FAX)
The materials in this message are private and may contain Protected Healthcare Information. If you are not the intended recipient, be advised that any unauthorized use, disclosure, copying or the taking of any action in reliance on the contents of this information is strictly prohibited=2E If you have received this email in error, please immediately notify the sender via telephone or return mail.
Histonet mailing list
<< Previous Message | Next Message >>