RE: [Histonet] vascular graft_sectioning difficulty
Can you provide a little more information about your processing, and a
little more detail about how the sections are crumbling (eg, tissue is
disaggregating, or the paraffin section is falling apart, etc)? We regularly
section formalin-fixed, paraffin-embedded vascular grafts (ePTFE) at 5-7µm,
and it's working for us. With a little more information, I might be able to
Tissue Genesis, Inc.
>From: "Dr. Raghul"
>Subject: [Histonet] vascular graft_sectioning difficulty
>Date: Tue, 14 Sep 2004 16:56:23 +0530
>We cut 5 µm thin sections of paraffin embedded vascular graft(Dacron) in
>a rotary microtome with disposable blades but the senior tech tells that
>the tissue sections are crumbling and forces us to go for thick sections
>around 10µm but again not with much success.
> We can understand that there is not enough tissue adhereing to the
>graft and we are literally cutting the graft. But did anyone had such
>problems sectioning vascular grafts. Any suggestions?
>Histopathology- implant biology division
>Srichitra tirunal institute of medical sciences and technology
>Trivandrum -695 012
>Histonet mailing list
FREE pop-up blocking with the new MSN Toolbar – get it now!
Histonet mailing list
<< Previous Message | Next Message >>