Re: herceptest antigen retrieval
|From:||Amos & Theresa <email@example.com>|
But not an FDA approved one :-)
Abizar Lakdawalla wrote:
> a pressure cooker would be a better and SAFER bet.
> Kappeler Andreas wrote:
> > Hi Marg
> > Welcome to the club! When we gave the manufacturer of the Herceptest a bad
> > time for their protocol (because it has been developed at sea level where
> > water boils as close to 100 C as possible) they were not very enthusiastic
> > about our comments (I'm stupid enough to live at a higher altitude). We've
> > tried almost everything under the sun using waterbaths, but were unable to
> > maintain 95 C for the buffer INSIDE the coplin jar, although we managed to
> > have this temperature for the waterbath, but that's not the point. Now, as
> > we do not live in FDA-country (even though I'm sure that the new president
> > of the USA - if there ever will be one - will within short declare FDA
> > rules to be the standard for the whole known world, i.e. the USA), we
> > modified the manufacturer's protocol and on an on-site inspection, the Dako
> > rep approved of our method - and a couple of months later asked for our
> > protocol ... Basically we are boiling the slides directly in the buffer,
> > using a Bunsen burner (which already gave me a couple of nasty comments on
> > the histonet on an earlier occasion). However, if you're living at too high
> > an altitude, where water boils at less than 95 C, our method will not solve
> > your problem. Insist with the manufacturer that they solve your problem (and
> > don't accept sending out the slides to a lab at sea level as a solution).
> > I'm including the protocol for you (but not the histonet) as a PDF file.
> > Good luck.
> > Andi Kappeler
> > Institute of Pathology, University of Bern, Switzerland
<< Previous Message | Next Message >>