Re: VIP-5 Owners OOPS
Sorry guys! Wipe the back of the bottle[metal area and O ring ]with
----- Original Message -----
From: "Melissa Jensen"
To: ; "Cynthia Favara"
Sent: Tuesday, January 14, 2003 7:09 PM
Subject: Re: VIP-5 Owners
> WE had the same problem last week and today.It took 4 of us to try to get
> that bottle out! That is nuts! Called in..Tech support said ...to remove
> bottle[lol]..than wipe the back the of the bottle [insert] with xylene and
> non lint gauze.Than grease the insert...Never use anything that is not
> free....They say the insert connector tends to dry out....I didn't take
> call so I have no clue what kind of grease they mean..I'm going to call
> tomorrow..Co worker found some in Chemistry and that's what we used
> today..may have to make this a monthly thing...Hope this helps...-----
> Original Message -----
> From: "Cynthia Favara"
> To: "'Gamble,Marilyn S'" ; "'Histonet'"
> Sent: Monday, January 13, 2003 5:42 PM
> Subject: RE: VIP-5 Owners
> > Marilyn,
> > I have had the VIP since Oct 02 . Have no processing problems but I have
> > three reagent bottles that I cannot remove. Sakura is sending a tech to
> > the problem.
> > Cynthia Favara
> > NIAID/NIH/RML/LPVD
> > 903 South 4th Street
> > Hamilton, MT 59840
> > 406-363-9317
> > -----Original Message-----
> > From: Gamble,Marilyn S [mailto:Marilyn.S.Gamble@kp.org]
> > Sent: Monday, January 13, 2003 3:44 PM
> > To: 'Histonet'
> > Subject: VIP-5 Owners
> > I've been off the Histonet for an extended period of time, so forgive me
> > this subject is a repeat.
> > We have had some problems with several VIP 5 processors, without error
> > showing. To determine if we have an employee problem instead of an
> > instrument problem, I'd like to check with other users of the new
> > instrument. I (There are no problems with the older versions of the
> > If you have had any issues with the instrument, would you send your
> > responses to me please.
> > Thank you.
> > Marilyn Gamble
> > Quality Assurance Coordinator, Cellular Pathology
> > Regional Reference Laboratory, Sherman Way
> > (818) 503-7029, tieline 397
> > E-mail: Marilyn.S.Gamble@kp.org
<< Previous Message | Next Message >>