[Histonet] problems with leica microtome
|From:||"Colegio de Histotecnologos" |
I read about the thik and thin sections problems. I have a 2125
microtome. Í´ve been having the same problem with mine.
I changed the leaf spring, the clamping lever and the clamps,
three times, it start
working fine and after a few moth it brokes, the leaf spring wear
down and start working bad again. Does any
one now if there is a recall for those knife holders?? Its wird
everyone is having the same problem. isn´t?
HT Anaeva Aleo
Las mejores tiendas, los precios mas bajos, entregas en todo el mundo,
YupiMSN Compras: Haz clic aquí...
Histonet mailing list
<< Previous Message | Next Message >>