RE: [Histonet] "empty" cryosections!
|From:||"Favara, Cynthia (NIH/NIAID)" |
Quick reply -
I frequently cool my knife with dry ice or with that freezing spray helps
Just need to keep trying sometimes.
903 South 4th Street
Hamilton, MT 59840
The information in this e-mail and any of its attachments is confidential
and may contain sensitive information. It should not be used by anyone who
is not the original intended recipient. If you have received this e-mail in
error please inform the sender and delete it from your mailbox or any other
storage devices. National Institute of Allergy and Infectious Diseases shall
not accept liability for any statements made that are sender's own and not
expressly made on behalf of the NIAID by one of its representatives
From: Joanne Whitehead [mailto:email@example.com]
Sent: Monday, October 03, 2005 2:04 AM
Subject: RE: [Histonet] "empty" cryosections!
First of all, many thanks to Gayle and everyone else who has offered
suggestions on freezing and sectioning.
I am using a Jung CM3000 cryostat with disposable blades. The machine
is showing it's age - it has needed several visits from the Leica
technicians lately. But I have just installed a new antiroll plate,
which is a bit of help. I have tried the brush technique also, but I
can see as the tissue comes across the knife it is already pulling
away from the OTC, so this is not an issue of the antiroll plate. I
hadn't considered variations in temperature within the machine - I
will check for that next time. I wondered if it would help if the
knife was colder than the tissue - anyone tried this? A couple of
pellets of dry ice on the knife holder perhaps?
I will definitely try embedding the tissue in OTC before freezing next
time, as well as infusing with OTC before freezing. But I wonder how
to make a suitable mold? Someone has suggested aluminum foil. I
ususally use re-useable plastic rings if the tissue is small enough,
or else cut a piece of the bulb from a disposable plastic pasteur
pipette for larger pieces, which I can just cut away after freezing.
Also I understand from the comments here that mixing dry ice directly
into the isopentane, rather than having the isopentance suspended
above liquid nitrogen will give a more consistent temperature for
snap freezing - another modification I will try next time.
I had already settled on -26C for sectioning, as suggested by Gayle. I
need to open up the intestine for my manipulations, so infusing the
tube directly with OTC is not an option for me. But I need sections
only 1 cm long, so it rolls up into a reasonably compact form. I have
been rolling with villi out so they will be spread apart better, but I
have also seen images rolled villi in - any preferences out there?
So thanks everyone, hopefully my next attempts at freezing &
sectioning will go much smoother. That brings me to my next problem,
which I will describe in a separate post :)
Histonet mailing list
Histonet mailing list
<< Previous Message | Next Message >>