This sounds like over-processing. If you are using a regular processing
schedule it is for sure. At Sacred Heart Medical Center in Spokane we used a
2 hour 45 minute program with 15 minutes in each station. We still soaked
the blocks in water before icing and cutting. Sometimes we had to soak for
the second or third level. We fixed in Bouin's. Also cutting with a slower
speed as the knife passed through the tissue helped. Hope this helps
Mickie Johnson, B.S., HTL(ASCP)
Mohs Histology Consulting Services, LLC
& Mohs Lab Staffing
& Mohs Suite Mohs Reporting Software
2507 S. Manito Blvd.
Spokane, WA 99203
Web: www.mohshistogyconsulting.com & www.mohslabstaffing.com &
This message is intended for the sole use of the addressee, and may contain
information that is privileged, confidential and exempt from disclosure
under applicable law. If you are not the addressee you are hereby notified
that you may not use, copy, disclose, or distribute to anyone the message or
any information contained in the message. If you have received this message
in error, please immediately advise the sender by reply email and delete
this message. Thank You.
[mailto:email@example.com] On Behalf Of Evans,
Sent: Thursday, November 13, 2008 12:46 PM
Subject: [Histonet] GI Biopsies
We are having difficulty with our GI biopsies having a lot of chatter on the
H & E. I think somehow they are getting over processed, but not sure....
has anyone had this problem or could suggest what might be causing this
Andria B Evans, HTL(ASCP)CM
1001 S. George Street
York, PA 17405
"You can learn a lot more from listening than you can from talking. Find
someone with whom you don't agree in the slightest and ask them to explain
themselves at length. Then take a seat, shut your mouth, and don't argue
back. It's physically impossible to listen with your mouth open." -John Moe
"Maturity is accepting imperfections."
This email may contain confidential health information that is legally
privileged. This information is intended for the use of the named
recipient(s). The authorized recipient of this information is prohibited
from disclosing this information to any party unless required to do so by
law or regulation and is required to destroy the information after its
stated need has been fulfilled. If you are not the intended recipient, you
are hereby notified that any disclosure, copying, distribution, or action
taken in reliance on the contents of this email is strictly prohibited. If
you receive this e-mail message in error, please notify the sender
immediately to arrange disposition of the information.
This e-mail has been scanned by MCI Managed Email Content Service, using
Skeptic(tm) technology powered by MessageLabs. For more information on MCI's
Managed Email Content Service, visit http://www.mci.com.
Histonet mailing list
No virus found in this incoming message.
Checked by AVG - http://www.avg.com
Version: 8.0.175 / Virus Database: 270.9.2/1783 - Release Date: 11/12/2008
Histonet mailing list