[Histonet] Stainng Problems-Ventana -- Benchmark
Re Robert Bates florida
Who manufactures your positive charged slide?
We had ER cases where the positive control on the top of the slide stained
and the a known positive patient case on the bottom half did not. Internal
control did not stain. Traced the problem to slide manufactor.
Manufactured failed to have an even distribution of charge on the entire
slide. Use Erie Slides
Windsor Regional Hospital
1995 Lens Avenue
519-254-5577 ext 52373
Histonet mailing list
<< Previous Message | Next Message >>