Re: [Histonet] zinc fixative
If your tissue is fixed in zinc formalin, the formulation should be with
zinc sulfate and not zinc chloride because zinc chloride is corrosive to
metal parts on enclosed tissue processors. In most instances zinc formalin
can be used on an enclosed processor again as long as it is made from zinc
sulfate. There is no need to follow with NB formalin on the processor.
Freida Carson has a reference to zinc formalin in her book Histotechnology
- A Self - Instruction Text.
Hope this helps.
Mari Ann Mailhiot BA HT ASCP
Leica Technical Assistance Center
800 248 0123 x7267
847 236 3063 fax
"PATRICE BOULIANNE" =20
Sent by: cc: =20
firstname.lastname@example.org Subject: [Histonet] zinc fixative =20
10/23/2003 11:13 AM =20
I want to know if after a fixation with zinc fixative(for IHC), we
can process the tissue in a regular formalin program (paraffin
impregnation)? Should I change the first bath (fromalin) for the zinc
fixative (with formaline). Is it corrosive for the processor? And should I
rince the tissue in tap water before the processing to release zinc
Thanks for your help!
Histopathotechnologist RT, Team Leader
Lab Pre-Clinical Research Intl. Inc.
450-973-2240 ext. 1625
This email has been scanned for all viruses by the MessageLabs Email
Security System. For more information on a proactive email security
service working around the clock, around the globe, visit
Histonet mailing list
<< Previous Message | Next Message >>