[Histonet] VIP Tissue Processor malfunctioning.....
Has anyone ever had a problem w/ their VIP processor pumping water into
the last paraffin bath or retort chamber? This has happened four or
five times in the last three months. It seems to always happen after I
change the reagents. Am I to blame?
P.S. What is the water for? Is its purpose to minimize fumes? I've
heard someone say it's used to pump reagents in and out....something
related to the vacuum and pressure, but this doesn't make sense to me...
Histonet mailing list
<< Previous Message | Next Message >>