Kathy Tedford-Hays HT (ASCP)
Technical Specialist, Histology Dept
(601)-968-3070 ext 7398
Baptist Medical Center
1225 North State Street
Jackson, MS 39202
i have the ventana special stainer and at first i was having the same
complaints from my pathologist concerning the light staining, level 4 is
the longest staining method that ventana has. on the same run some slides
would be dark and then the very next one would be too light. ventana tech
support came in and made a few suggestions, 1. take the kit out of the
refrig for at least 30 mins before the run at room temp. 2. invert the
kits to shake up the silver on the bottom. 3. do 2 sets of cleaning
reagent through the tubing. 4. make sure all of your microtomes are
cutting on 4 microns. these extra sets have worked for us. hope this helps.
05/09/03 03:30 AM Subject: Warthin-Starry
Good Morning Everyone,
Is there ae easy method for W-S, silly question. We use the Benchmark
Steiner kit at Level 4
and the Pathologists say it is still too light. Can the intensity be
increased off the machine?
Thanks for your help
UIC Medical Center
STOP MORE SPAM with the new MSN 8 and get 2 months FREE*
This internet email has been certified virus free by BHS Anti-Virus system.
The information contained in this email is confidential and is intended for the recipient only. Do not forward this email without permission from its originator.
<< Previous Message | Next Message >>