RE: Pipette Calibration
|From:||"Brennan, Lynn" <LBrennan@coh.org>|
> -----Original Message-----
> From: Nick_Madary@hgsi.com@COHNOTES
> Sent: Thursday, May 31, 2001 8:59 AM
> To: firstname.lastname@example.org
> Subject: Pipette Calibration
> Does anyone know the acceptable range of "weights of water" used when
> calibrating pipettes? For example the weight of 500 ul should be.....
> I am sure the reference is right in front of me but poor ventilation has
> destroyed many neurons.
1 ml of water weighs 1 gram. 1 ul of water would weigh 0.001 gram
or 1 mg or 500 ul would weigh 0.5 grams.
<< Previous Message | Next Message >>