Re: [Histonet] Target retrieval solution
Sorry... I was writing solely about Target Retrieval solution, and not about
the Decloaker in particular, in my previous email. I perform HIER in a
microwave, where I can set any time I want. The 10-15-20 minutes pertains
to the heating time only. There is also the 20 minute cooldown in solution
afterward, independent of the timing of the heating length.
I also have a Biocare Decloaker which I use for CWD/Scrapie tissues only.
The SP1 is the heating time length at high pressure; once the decloaker
reaches the set temperature/pressure, the time countdown starts; the time
can be adjusted up or down. The SP2 is the depressurizing step. My
decloaker is set for 20 minutes at SP1, and 25 minutes at SP2.
U of MN
----- Original Message -----
From: "Katri Tuomala"
To: "Jan Shivers" ; "Larry Woody"
Sent: Tuesday, March 29, 2005 12:21 PM
Subject: Re: [Histonet] Target retrieval solution
> Jan and everyone else using the decloaker:
> The Biocare decloaker has two stages: SP1 and SP2. When you say running
> is 10, 15 or 20 minutes, is it total of two stages or just the SP2 stage?
> I remember correctly SP1 can only be set maximum 5 minutes. Could someone
> clarify this for me?
> Katri Tuomala
> Hamilton, Ontario, Canada
Histonet mailing list
<< Previous Message | Next Message >>