|From:||"Blair, Keith R Mr IACH" |
i'm using a ventana renaissance. i've experienced none of those problems. it appears to be very reliable so far. we/ve only had it 2 years.kblair ht/htl ft riley ks
From: Vickroy, Jim [mailto:Vickroy.Jim@mhsil.com]
Sent: Friday, March 08, 2002 1:20 PM
Subject: Ventana Tissue Processor
Lately we have experienced something that I have not seen in 27 yrs of
We have a Ventana tissue processor (1 year old). It has had numerous
software updates. The other day we experienced a nightmare. Our tissue
came out with a cooked appearance. There were no alarms that went off and a
run report says no problems.
The tissue was barely salvageable but was very hard and when looking at it
under the scope appeared cooked. I have discovered that another Ventana
user has had the same problem only their instrument has been doing this
three days in a row. We have eliminated reagent problems, etc. Has anyone
else experienced these problems with the Ventana Processor.
Before this occurrence the instrument ran fine except occasionally there was
some blocks that had fluid in the center of them instead of paraffin. The
company is looking at other possible reasons and I want to see if anyone is
having the same problem.