RE: Benchmark error code
Yes we are experiencing the EZ prep error code. Moreso now in the last 2-3
weeks. We just got some new valves in and we changed all the EZ prep valves.
This seems to have solved the problem (at least for now!)
Our area sales rep says other places are also experiencing the same problem,
so it may be something in the latest batch of EZ prep solution.
Toronto General Hospital
> -----Original Message-----
> From: Lynn McKee [SMTP:email@example.com]
> Sent: Wednesday, July 24, 2002 12:01 PM
> To: firstname.lastname@example.org
> Subject: Benchmark error code
> Dear Histonetters,
> We're curious to see if anyone with a Benchmark experienced the error
> code "EZ Prep resevoir fails to charge" ?
> Lynn McKee
> University of Cincinnati
<< Previous Message | Next Message >>