I've been off the Histonet for an extended period of time, so forgive me if this subject is a repeat.
We have had some problems with several VIP 5 processors, without error codes showing. To determine if we have an employee problem instead of an instrument problem, I'd like to check with other users of the new instrument. I (There are no problems with the older versions of the VIP). If you have had any issues with the instrument, would you send your responses to me please.
Quality Assurance Coordinator, Cellular Pathology
Regional Reference Laboratory, Sherman Way
(818) 503-7029, tieline 397
<< Previous Message | Next Message >>