RE: [Histonet] processing
I infiltrate my GI biopsies with a low melting point paraffin (Paraplast Xtra) at 58*C on my VIP processors. GI biopsies easily become friable at paraffin temperatures of 62*C and over. Ask her to check the temps of her paraffins between the two processors. Also, GI biopsies seem to be more challanging to cut after fresh dehydrating alcohol solutions are added to the processors. They respond well to graduated alcohols from 70%, 80%, 95%, 100% to the clearant. I do not use the Ventana processor, but I do know it does require some fine tuning to program the dehydrating alcohols to match those on the VIP.
Eric C. Kellar
Quest Diagnostics Inc. Miami
[mailto:firstname.lastname@example.org]On Behalf Of Webb,
Sent: Wednesday, February 22, 2006 12:12 PM
Subject: [Histonet] processing
I am submitting this for a colleague at another institution who has a
concern over GI biopsies. They have 2 processors they use, one VIP and
one Renaissance, both programmed with the same times and the same
reagents at each station. They use recycled formalin. Their
pathologists are complaining that the Gi biopsies that come off of the
Renaissance processor are not of good quality, chattered, drier, etc.
and the problems are not seen on the slides of the biopsies from the
other (VIP) processor. They had a vendor out to check out the processor
in question, inasmuch as they could be getting carry over of reagents,
etc. and all checked out ok. Any suggestions as to the cause of their
problem? Thanks ahead of time from a lead tech who is pulling out her
hair!!!! I told her to "hold on", as my fellow histonetters would come
up with something!!
This e-mail and any files transmitted with it are confidential and are intended solely for the use of the individual or entity to whom they are addressed. If you are not the intended recipient or the individual responsible for delivering the e-mail to the intended recipient, please be advised that you have received this e-mail in error and that any use, dissemination, forwarding, printing, or copying of this e-mail is strictly prohibited.
If you have received this e-mail in error, please immediately notify the HealthPartners Support Center by telephone at (952) 967-6600. You will be reimbursed for reasonable costs incurred in notifying us.
Histonet mailing list
The contents of this message, together with any attachments, are intended only for the use of the person(s) to which they are addressed and may contain confidential and/or privileged information. Further, any medical information herein is confidential and protected by law. It is unlawful for unauthorized persons to use, review, copy, disclose, or disseminate confidential medical information. If you are not the intended recipient, immediately advise the sender and delete this message and any attachments. Any distribution, or copying of this message, or any attachment, is prohibited.
Histonet mailing list
<< Previous Message | Next Message >>