Re: [Histonet] Superfrost plus slides
We've have a similar problem on occasions with Surgipath slides as well. We also have the DAKO Autostainer and use DAKO diluent and detection reagents.
> from: Mike Bromley
> date: Tue, 23 Dec 2003 11:22:07
> to: email@example.com, firstname.lastname@example.org, email@example.com, firstname.lastname@example.org, email@example.com
> subject: Re: [Histonet] Superfrost plus slides
> Hi All
> We have had a problem with a batch of Menzel Superfrost Plus slides
> (3451272). They have been made in a way that makes them more hydrophobic
> than usual.
> The problem may not have been noticed by others because it only occurs under
> certain circumstances, however the consequences can be extremely serious.
> The circumstances are:
> 1. Use of a "flatbed" stainer where the slides are held horizontally
> 2. Use of a bad batch of slides
> 3. Use of these slides for antibodies which do not require pressure
> cooking, this usually means enzyme treatment or no treatment (sometimes even
> when heat treated in a 95-99* water bath as in the Herceptest) Pressure
> cooking the slides seems to overcome the hydrophobic nature of the "bad"
> Wash buffer usually spreads okay on the slides, but when the reagent is
> dripped onto the slide it fails to spread evenly. This is usually not a
> problem if the section is a single piece of tissue and it is hit by the
> drops however if the section is composed of small pieces then some of these
> can be missed. Hence your controls may be fine, because they are often a
> single piece of a reasonable size, but you may miss a critical piece of
> I advise everyone who uses a flatbed horizontal stainer for sections not
> requiring pressure cook antigen retrieval to watch the addition of the
> dropped reagents. This problem is due to batch variation and may be slight
> or extreme depending on the batch.
> Other users have noted problems with Superfrost Plus slides before (see
> histonet archives) We use Superfrost Plus slides because they are better
> than home made APES slides especially for our slides that are pressure
> cooked in high pH solution.
> All our reagents are from Dako and the Autostainer protocol is followed.
> Best Wishes
> Mike Bromley
> Chief Biomedical Scientist
> Dumfries & Galloway Royal Infirmary
> Scotland, UK
> > This e-mail and any files transmitted with it are private and intended
> > solely for the use of the individual or entity to whom they are addressed.
> > If you have received this e-mail in error please return it to the address
> > it
> > came from telling them it is not for you and then delete it from your
> > system.
Histonet mailing list
<< Previous Message | Next Message >>