RE: [Histonet] Steve Slap Please Read
We process placentas daily in our Milestone RHS1 unit without any problems.
I can send you the protocol via fax if you will send me a number or by
e-mail if your system will allow you to receive attachments.
Histology Lab Manager
Harris Methodist Fort Worth, Tx
From: Steven E. Slap [mailto:firstname.lastname@example.org]
Sent: Monday, January 05, 2004 9:24 AM
To: DMBCMP@aol.com; email@example.com
Subject: Re: [Histonet] Steve Slap Please Read
Hi Dannie & HistoNetters
Happy 2004 to All
First, to address the side issue of finding histotechs on the
Internet. Please don't forget that many of us (myself included) can
be e-mailed directly from "The Directory of Histotechs on the Web" at
http://www.histology.to. If you are not yet listed, but want tobe,
there is a submission form available, or you can just send your
listing to me at mailto:firstname.lastname@example.org.
Now, forgive me a brief plea. Peggy Wenk and I have maintained this
web site, "The Histotech's Home Page", as a volunteer effort, for
some years now. Over the years, we have been aided by the generous
financial support of several commercial sponsors, but we are now
down to one sponsor (thank you, Pacific Southwest Lab Supplies!!!),
and this is not nearly enough to pay the bills associated with
maintaining the site. As I am now self-employed, I cannot afford to
support the site on my own. I hope that several companies can step
forward in 2004 to help out and keep this valuable resource available.
Now to Dannie's question.
The purpose of the vacuum step in microwave histoprocessing is to
remove all residual isopropanol from the tissue. The problem with
the placentas, I think, is twofold. First, there may still be
residual water in the tissue (try extending the isopropanol step).
Second, there is a very large amount of alcohol being carried over
within the tissue into the vacuum step into the paraffin. As the
isopropanol flash evaporates, it is causing a "volcano" effect in the
paraffin, and paraffin is splattering up into the vacuum. Both of
these problems appear only with very large tissues, and would not be
seen with biopsies. In addition to extending the isopropanol step, I
would make sure that the paraffin is at the correct temperature at
the start of the cycle, and is not overfilled as first steps to
resolving the problem.
Dannie- thanks for the compliment, and feel free to contact me directly.
At 4:00 PM -0500 1/1/04, DMBCMP@aol.com wrote:
>No matter what, you are still the microwave specialist and I need your
>if you would be so gracious as to reply.
>Our Milestone Renaissance is being used daily for a run of placenta blocks.
>(I know bloody specimens are not highly recommended for microwave.)
>Do you have
>any idea why paraffin is being sucked into the vacuum tube? This is
>happening at various times....not necessarily when it has a full
>load. Our first run
>of biopsies is fine. The placentas are on the second run. The paraffin is
>to temp @ 80 C. We were wondering if the problem is simply because they
>bloody tissue and we just need to change the paraffin if we wish to use the
>microwave for this application. What do you think? I sure would
>expert knowledge on this.
>NOTE: Anyone else who has any ideas (Tim Morken, during your CDC days?)
> you are most welcome to jump in.
>Steve, I hope you had a good holiday. Happy 2004!!!!
>Same to all you people around the world who read and learn from HistoNet.
>HAPPY NEW YEAR!!!
>Dannie Blake, HT (ASCP)
>Sierra Pathology, Fresno Community Hospital
>Histonet mailing list
Histonet mailing list
The information contained in this message and any attachments is intended only for the use of the individual or entity to which it is addressed, and may contain information that is PRIVILEGED, CONFIDENTIAL, and exempt from disclosure under applicable law. If you have received this message in error, you are prohibited from copying, distributing, or using the information. Please contact the sender immediately by return e-mail and delete the original message from your system.
Histonet mailing list
<< Previous Message | Next Message >>