[Histonet] Problem with VIP 5
We are having a problem with our VIP 5 having a strong formalin odor at the
end of the run when we open the retort to remove the tissues. There is also
a large amount of fluid-formalin?? on the lid and in the chamber.
Has anyone else experienced this? I have called Sakura and they are sending
a field service tech but I was curious if anyone has had this problem.
Whenever I call a vendor it seems they always say, well, we haven't had
THIS problem before.
Jill Songer HT (ASCP)
Virginia-Maryland Regional College of Veterinary Medicine
Blacksburg, VA 24061-0443
Histonet mailing list
<< Previous Message | Next Message >>