Re: [Histonet] Ventana Tissue Processor
I have had this problem before on two different type of processors. I have
researched and researched what could have happened....chemicals were placed
appropriately, heat is applied only on the paraffin stations, processing times
are standard...the only thing that I have came up with is that if you are
processing very small delicate tissue directly after a complete change on the
processor, the tissue is actually getting sucked dry by dehydration. Look at your
maintenance records again and see if the processor had just been completely
I just recently attended a seminar "The chemistry of processing and staining"
and the lecturer referred to it as removal of the bonded water ...the bonded
water is located at the cellular level (within the cell) and you do not want
to remove this. If the tissue is becoming over dehydrated it is removing this
bonded water as well as the free water and is causing the tissue to shrivel up
and in essence probably getting cooked.
I have resorted to using sponges with tiny fragile tissue and doing partial
changes if I know that I will be processing delicate tissue and so far this has
Another helpful hint in the event you would be cutting these tissue blocks is
to soak the tissue blocks after faced in a cold glycerin solution...it will
help tremendously with sectioning...although microscopically the outer edges
will look "cooked". Hope this helps.
Histonet mailing list
<< Previous Message | Next Message >>